A Brief Explanation of Phase
To demonstrate what phase really is let's first examine a system with ideal phase response passing a square wave. Look at this with the principle
of superposition in mind. The square wave is actually the sum of decreasing amounts of the odd harmonics of the fundamental. All of these harmonics are in
phase with the fundamental (and with each other). To demonstrate this let's pick a reference point, say the positive going zero crossing of the fundamental as
viewed in the time domain (an oscilloscope trace). All of these harmonics will also have a positive going zero crossing at this reference point in the cycle of
Square wave & its first 4 components (1st, 3rd, 5th & 7th harmonics)
If phase distortion is introduced by a system with non-ideal phase response, all of the harmonics are no longer in phase with the fundamental. In other
words, at the fundamentals positive going zero crossing some of the harmonics will no longer be at their positive going zero crossing. As an example, given
a 1 kHz square wave, if there is -90 degrees of phase shift at 5 kHz (relative to 1 kHz) then the 5th harmonic that makes up the square wave will have a
minima, not a positive going zero crossing, at the reference point in the cycle of the fundamental.
Square wave with phase shift as described above
Let's now turn our attention to the frequency domain. Looking at a TEF/Smaart plot of phase vs. freq. for our system with ideal phase response, after having
removed any excess delay, we would see a straight horizontal line. This would be indicating to us that all frequencies have 0 degrees phase shift relative to
each other. If we now examine the system with non-ideal phase response, say a loudspeaker, we would see a curve that starts out at X degrees at some low
frequency. As frequency increases the overall trend of the curve is one of decreasing phase. At 1 kHz the phase curve may indicate -50 degrees while
at 5 kHz it indicates -140 degrees, in keeping with our example above.
An important point to keep in mind is that the 0 degrees phase line on a TEF/Smaart plot and the zero crossing on a oscilloscope trace are not necessarily the same thing.
Also note that the setting for the receive delay on the TEF/Smaart to remove the excess delay has only one value for a given measurement. Any "tweaking" of
this value results in one of two consequences:
- Receive delay too small - not all of the excess delay has been removed and the phase plot will show too much downward slope;
- Receive delay too large - the TEF/Smaart measurement system is "listening" after the initial arrival of energy (typically HF) which results in the phase plot
turning up in the HF region.
The latter is indicative of an acausal system in which energy comes out before it is put in. Both of these conditions can allow for 0 degrees phase at a
given frequency on the curve.